Information on pending and completed system maintenance issues
4/17/2021 -- Problem with ICRON box (USB extender) being intermittent. Replaced power supply and several cables. Attached spade lugs to wiring at Digital Logger switch. APC UPS firmware update failed due to bad information from APC tech. New cable ordered. Update firmware on Digital-Logger switches. Analyze shutter battery -- needs replacement (on order). Installed new keyboard on computer.
3/8/2021 -- The telescope system has been out of service for a while. There was a problem with a bearing in the RA drivetrain. This problem has been fixed and the telescope is operational. There is always a maintenance list and an upgrade list of things to do.
12/20/2020 -- The Madrona Peak Observatory telescope has been in operation for most of 2020, without intervention from the 'maintenance crew'. In December , 2020 the telescope went out of service pending resolution of a problem with the drive system. The telescope will not move in RA. Work is in progress to identify and fix the problem.
1/11/2020 -- Installed primary imaging camera (now repaired) on the telescope. Cleaned the field flattener lens between the camera and the 'scope. Replaced wiring to camera from power supply to reduce voltage drop. Camera draws significant current when cooling, and the rewiring keeps the voltage to the camera above the minimum requirement (reduces resistive losses). Changed power source for camera from rack supply to the power supply provided with the camera.
11/23/19 -- The primary imaging camera has been returned after repairs (10 months). Installation is pending. The ICRON box (USB extender) that provides connectivity between the imaging camera and the computer (and other connections) has been replaced due to failure in the original unit. The new ICRON required a different power supply, which was installed. To avoid ground loops the 12 VDC power and the ICRON power grounds were connected together and to the telescope frame. The secondary imaging camera, which has been in place since February, 2019 has functioned intermittently. Cleaning the electrical connections seems to have helped. Yet another ICRON USB extender to transmit weather data from the console to the computer is pending installation. Updated the Acronis backup software to latest version.
9/8/19 -- After a bit of work the communications between the Obs and the Wx computers to their associated UPS ethernet cards is working. This allows for automated shutdown of the system in the event of a power outage. System is unavailable, though, due to no communications between the imaging camera and the Obs computer.
7/28/19 -- Resolved problem with system not returning to 'ready' state after a power outage. A setting in the Digital-Logger switches had to be changed to force a return to previous conditions. The imaging camera was not communicating with the imaging software. This was the result of a failure of a USB port connected to the computer. Moving the camera to a different USB port solved the problem. Installed ethernet cards in the APC UPS units. Configured the cards to be active on the ranch network. Successfully tested a new (home grown) dome driver to correct problems associated with previous dome driver. Resolved problem with connection of USB cable between UPS and computers.
2/22/19 -- Installed two new UPS units (and new management system), a second Digital-Loggers power switch, new weather sensors (temp, humidity, wind, etc.). Configured new Unifi network switch. Checked function of observatory. Ok, but Apogee Aspen camera had moisture on CCD; replaced with backup camera. Aspen camera returned to manufacturer for repair. Backup camera installed and tested, but new calibration images need to be taken. Surveillance camera previously installed. Checked computer backups to confirm that they are working. Davis weather console not working; new one ordered.
12/17/18 -- Some of the problems at the observatory are resolved. Internet communications to the observatory computer is now available. A trip is scheduled to install new remote power switches and a new surveillance camera. We expect this effort will return the system to full functionality.
10/29/18 -- Plan to assess the situation at the observatory in a couple of weeks followed by remedial work (if needed).
10/12/18 -- Observatory maintenance needed. System computer has malfunctioned along with communications to the observatory. A repair trip is being scheduled.
5/17/18 -- Cleaned cloud sensor. Checked clearance of mirror cover and camera slot. Resolved problem with guider camera (not being seen by computer). O/S update to weather computer resolved problem with restart.
1/6/18 -- Replaced hard drive in weather monitoring computer with 500 GB SSD. Resolved problems with slow booting of computer and failure to restart after issuing 'restart' command. Updated BIOS, reset boot on power connected. Resolved problem with 'Share' used to communicate weather information.
1/6/18 -- Tested main imaging camera USB connection to determine if USB and ethernet cables can be connected simultaneously. They cannot.
1/6/18 -- Cleaned lens of all-sky camera.
1/6/18 -- Resolved problem with UPS communication with computer. (New cable)
1/6/18 -- Identified problem with monitor in observatory not working.
1/6/18 -- Reconfigured power switches to isolate ethernet switch.
1/6/19 -- Tested emergency shutdown script (more work needed)
12/1/17 -- Pending work -- replace hard drive in weather monitor computer
3/11/17 -- Re-install original camera Apogee Aspen after repair. Recalibrate guide camera
1/28/17 -- Create wiring block diagrams for observatory control systems
1/28/17 -- Recalibrate autoguider
1/28/17 -- Update focuser 'home' point
1/28/17 -- Clean filter on observatory air conditioner
1/27/17 -- Resolve network communications problem to observatory
1/27/17 -- Recalibrate autoguider.
1/27/17 -- Install new batteries in UPS units. Enable UPS monitoring.
12/9/16 -- Remove Apogee Aspen camera from telescope. Send to ANDOR for repair of leak in 'window' above imager chip. Install backup camera -- Apogee Alta
11/19/16 -- Install backup disks.
11/19/16 -- Acquire external disks for backups
11/19/16 -- Fix frost on imager problem -- camera returned to vendor for repair
11/19/16 -- Resolve problem with reboot on power up of observatory computer
11/19/16 -- Install new power supply for observatory WiFi access point
11/19/16 -- Photos for documentation of existing system configuration
11/19/16 -- Learn about ACP scheduler -- executed successfully
11/18/16 -- Activate pointing model for system
11/13/16 -- Fix all sky camera upload problem
11/13/16 -- Upgrade SkyX to version 10.5
11/12/16 -- First run of telescope by current operator; image of Ring Nebula taken. Pointing good but did not include pointing model.